||XRF Technologies for Measuring Trace Elements in Soil and Sediment: Innov-X XT400 Series XRF Analyzer. Innovative Technology Verification Report.
||Tetra Tech Environmental Management, Inc., Cincinnati, OH.;Environmental Protection Agency, Las Vegas, NV. National Exposure Research Lab.
X-ray fluorescence analyser ;
Cost data ;
Innov-X XT400 ;
Tables (Data) ;
Trace elements ;
Innovative technology ;
XRF instruments ;
Superfund Innovative Technology Evaluation (SITE) ;
X-ray Fluorescence (XRF)
||Most EPA libraries have a fiche copy filed under the call number shown. Check with individual libraries about paper copy.
||The Innov-X XT400 Series (XT400) x-ray fluorescence (XRF) analyzer was demonstrated under the U.S. Environmental Protection Agency (EPA) Superfund Innovative Technology Evaluation (SITE) Program. The field portion of the demonstration was conducted in January 2005 at the Kennedy Athletic, Recreational and Social Park (KARS) at Kennedy Space Center on Merritt Island, Florida. The demonstration was designed to collect reliable performance and cost data for the XT400 analyzer and seven other commercially available XRF instruments for measuring trace elements in soil and sediment. The performance and cost data were evaluated to document the relative performance of each XRF instrument.
||Sponsored by Environmental Protection Agency, Las Vegas, NV. National Exposure Research Lab.
||Order this product from NTIS by: phone at 1-800-553-NTIS (U.S. customers); (703)605-6000 (other countries); fax at (703)605-6900; and email at email@example.com. NTIS is located at 5285 Port Royal Road, Springfield, VA, 22161, USA.
||48E; 51F; 68; 48B
||PC A11/MF A03