||FT-IR Open-Path Monitoring Guidance Document.
Russwurm, G. M. ;
Childers, J. W. ;
||ManTech Environmental Technology, Inc., Research Triangle Park, NC.;Environmental Protection Agency, Research Triangle Park, NC. National Exposure Research Lab.
Fourier transform spectrometers ;
Infrared spectrometers ;
Michelson interferometers ;
Air pollution monitoring ;
Infrared spectroscopy ;
Infrared spectra ;
Water vapor ;
Quality control ;
Quality assurance ;
Performance evaluation ;
Open path monitors ;
FT-IR(Fourier Transform Infrared) ;
||Most EPA libraries have a fiche copy filed under the call number shown. Check with individual libraries about paper copy.
||This document describes the components of FT-IR monitors and is intended to provide guidance forthe FT-IR operator in field monitoring applications. A brief discussion of the FT-IR open-path monitor and its function is given in Chapter 2, along with a more in-depth description of the various components of the sensor. Chapter 3 includes the preliminary procedures for setting up the FT-IR instrumentation for monitoring. Chapter 4 is a discussion of background spectra, and Chapter 5 is a discussion of water vapor spectra. Chapter 6 presents guidance on how to set up the monitoring instruments within the physical constraints of a site. Chapter 7 presents experimental data that illustrate the effect of resolution and related parameters on the spectral data. Chapter 8 contains quality control and quality assurance guidelines, incorporating portions of an approved quality assurance plan, and includes selected QA data we collected over a recent one-year period.
||Sponsored by Environmental Protection Agency, Research Triangle Park, NC. National Exposure Research Lab.
||PC A09/MF A02