||Innovative technology verification report : Oxford ED2000 XRF Analyzer /
||Tetra Tech EM Inc.; Tetra Tech Environmental Management, Inc., Cincinnati, OH.; Environmental Protection Agency, Las Vegas, NV. National Exposure Research Lab.
|| U.S. Environmental Protection Agency, Office of Research and Development, National Exposure Research Laboratory, Environmental Sciences Division, Characterization and Monitoring Branch,
X-ray fluorescence analyzer ;
Cost data ;
Oxford ED 2000 ;
Tables (Data) ;
Trace elements ;
Innovative technology ;
XRF instruments ;
Superfund Innovative Technology Evaluation (SITE) ;
X-ray fluorescence (XRF)
||Most EPA libraries have a fiche copy filed under the call number shown. Check with individual libraries about paper copy.
||xiv, 73,  pages : color illustrations, charts (some color) ; 28 cm
The Oxford ED2000 x-ray fluorescence (XRF) analyzer was demonstrated under the U.S. Environmental Protection Agency (EPA) Superfund Innovative Technology Evaluation (SITE) Program. The field portion of the demonstration was conducted in January 2005 at the Kennedy Athletic, Recreational and Social Park (KARS) at Kennedy Space Center on Merritt Island, Florida. The demonstration was designed to collect reliable performance and cost data for the ED2000 analyzer and seven other commercially available XRF instruments for measuring trace elements in soil and sediment. The performance and cost data were evaluated to document the relative performance of each XRF instrument. This innovative technology verification report describes the objectives and the results of that evaluation and serves to verify the performance and cost of the ED2000 analyzer. Separate reports have been prepared for the other XRF instruments that were evaluated as part of the demonstration.
"EPA 540-R-06-007." "April 2006." Cover title. "Contract no. 68-C-00-181, task order no. 42." Includes bibliographical references (page 73).