||Environmental Technology Verification Report: Man-Tech Associates Inc., TitraSip (Trade Name) SA System Continuous Multi-Parameter Water Quality Monitor.
||Battelle, Columbus, OH.;National Risk Management Research Lab., Cincinnati, OH.
Water pollution sampling;
Environmental technology verification(ETV) program;
Man-Tech Associates Inc
||Most EPA libraries have a fiche copy filed under the call number shown. Check with individual libraries about paper copy.
||The Advanced Monitoring Systems (AMS) Center, one of six technology areas under ETV, is operated by Battelle in cooperation with EPA's National Exposure Research Laboratory. The AMS Center evaluated the performance of the Man-Tech Associates Inc. TitraSip (Trade Name) SA (Stand-Alone) System in continuously measuring total chlorine, temperature, conductivity, pH, total alkalinity, and turbidity in drinking water. This verification statement provides a summary of the test results. The performance of the TitraSip (Trade Name) was assessed in terms of its accuracy, response to injected contaminants, inter-unit reproducibility, ease of use, and data acquisition. The verification test was conducted between August 9 and October 28, 1004 and consisted of three stages, each designed to evaluate a particular performance characteristic of the TitraSip (Trade Name). All three stages of the test were conducted using a recirculating pipe loop at the U.S. EPA's Test and Evaluation Facility in Cincinnati, Ohio.
||See also PB2006-104806. Sponsored by National Risk Management Research Lab., Cincinnati, OH.
||Order this product from NTIS by: phone at 1-800-553-NTIS (U.S. customers); (703)605-6000 (other countries); fax at (703)605-6900; and email at email@example.com. NTIS is located at 5285 Port Royal Road, Springfield, VA, 22161, USA.
||68D; 50B; 99B; 91A; 43F